How to Use Pendulum Impact Testing Machine?
Pendulum impact testing machine (also known as smart pendulum impact tester) is a kind of electrical test equipment, which can precisely detects the impact properties of plastics, film, paper, composite membranes, foils and other materials. The equipment use electronic test, high degree of automation, equipped with professional software, which can process curve analysis, data storage, and contrast for the test data. Pendulum impact test machine technical features: 1. Testing machine is a single supporting column structure, cantilevered hanging pendulum mode, the u type pendulum hammer. 2. The hand control box is used to control the entire impact Women UGG Boots On Sale Classic test procedure. 3. Select ordinary pendulum impact testing machine, digital display pendulum impact testing machines and computer-controlled pendulum impact test equipment according to needs. 4. Specific impact can choose to read by liquid crystal display, or directly from the computer screen and intuitive reading. 5. The impact knife is installed and fixed with a screw,
easy replacement. 6. The host is equipped with a safety pin, and equipped Women’s UGGs On Sale with a safety net. Pendulum impact test machine usage: 1. Check whether testing machine is in a normal state, the moving parts and fasteners must be secure. 2. Determine and install the pendulum with appropriate range according to the sample material requirements, pay attention that the connection screws should be securely fastened. 3. Check and adjust the relative position of the specimen bearing and shock pendulum. Bearing jaw span should be 40 mm, jaw span and position adjustment is based on the pendulum blade, using a template to proceed. At the time of adjustment, loosen the screws, a little span of two jaws some boilerplate flat on the two upper jaw, lower the blade into the top of the pendulum and make a model on the 300 V-notch, the hand of the two jaws squeeze inward rope until the top tight template, and then tighten Discount UGG Boots For Women the screws so that the sample holder is pressed solid and reliable. 4. Check and adjust the dial needle When the pendulum is free at plumb position,
steering pointer to the maximum reading, adjust the dial needle pointer above the plane and try to be close with a small column, then rotate fastening screws on the dial needle. 5. Installing the sample First control the impact pendulum on the stand, place the sample to make it back to grooving blade, flat and brushed tight jaw in two bearings, and align with the alignment plate, so that the notch on the specimen is just located in the middle of the span,
the impact test can be carried out after removing the jaw blade plate. 6. Impact Test You can start by pressing the “back off” on the control box to “impact”. It should be reminded that, users should periodically service and UGG Sale Boots maintain the equipment to ensure the test accuracy and prolong equipment life.
For more information kindly visit electricaltest.cn/